Payment & Shipping Terms:
|Transportation:||Air Or Sea||Color:||Silver Chrome|
Cobalt Chrome Dental Casting Alloys for Frames or Partial Denture
Our cobalt chrome dental casting alloys enable to obtain smooth and compact surfaces and reduce the oxide formation. It has a strong resistance to traction and excellent workability. They are a perfect economical alternative to Gold & Silver based alloys (precious) without compromising its performance thanks to Willett's alloy mechanical properties and biocompatibility.
This packaging of 25KG is specially made for the Wholesale purpose.
Please let our sales representative know your packaging preference. We are also able to provide retail packaging alongside with the shipment.
|Melting Point||1100-1200 Celsius|
(1). Superior mechanical performance castability, toughness.
(2). Corrosion resistance, wear resistance, high intensity, great hardness.
(3). Nickle-free guaranteed biocompatibility.
What is the perfect temperature to obtain the best result when casting?
The casting temperature should always maintain 100~150℃ higher than liquidus temperature.
What is the perfect timing to obtain the best result when casting?
While casting Nickel Chrome alloys that don't contain Beryllium, the surface of the melting alloys will appear a layer of shadow. Once the shadow disappears and the edges of the alloy have become rounded (melted until it becomes a sphere shape is not necessary) will be the best timing for casting. Perfect timing for casting Cobalt Chrome alloy is to heat it till the surface membrane perforated and a spark of bright light start to appear.
Cobalt Chrome casting piece is incomplete, what are the causes?
1. Casting ring is not heated evenly: While casting Cobalt alloys, casting ring must heat up gradually and slowly before reaches 300℃. Let it sit in the oven in between 300℃ and 600℃ for 10-30 minutes. Once it heats up to the desired temperature, maintain the same temperature for at least 30 minutes before casting.
2. Casting temperature is below/above alloy's requirement.
3. Strub designed improperly: For example, Strub is too narrow. We strongly recommend the Strub diameter consist of 5mm or more.
4. The amount of alloy is not enough for the casting piece or the centrifugal force of casting pressure is not enough.
5. Casting piece has not avoided the heat center.